Our laboratory, on the recommendation of your specialist, purchased a small-sized microohmmeter MIKO-10 manufactured by SKB EP, LLC. Using the instrument leaves a positive feeling. We especially want to highlight its compactness, ease of use and performance at low temperatures. Not every instrument can brag about it. The instrument case is assembled efficiently, there are no backlashes and creaks. The settings menu is intuitive, autonomy is good, accuracy and measurement speed are also at a high level. The insulation of the measuring cables ensures that they remain flexible even at low ambient temperatures.
As for the minuses, the connectors stick and give a little inconvenience when plugging in or unplugging, it would be more convenient make connectors like "banana".
[Manufacturer note]: The fastening implemented in MIKO-10 microohmmeter is designed for reliable fixation of test cables to the device that excludes circuit break and ensures tight contact of cables with device inputs, which cannot be guaranteed in the presence of "banana" type connectors.
We also would like test cables with loaded needle-type contacts to be in paired with the instrument, not as additional equipment.
[Manufacturer note]: Standard complete set of MIKO-10 microohmmeter includes only one test cable (item code: SKB042.04.00.000), since the functionality of this cable is sufficient for most measurements. In our opinion, it is a universal cable for making measurements with MIKO-10. Test cables with spring loaded needle-type contacts are purchased depending on the measurement object requirements where a standard cable cannot be used.
Our laboratory recommends the MIKO-10 to purchase.